[Introduction] [Signal arithmetic] [Signals and noise] [Smoothing] [Differentiation] [Peak Sharpening] [Harmonic analysis] [Fourier convolution] [Fourier deconvolution] [Fourier filter] [Wavelets] [Peak area measurement] [Linear Least Squares] [Multicomponent Spectroscopy] [Iterative Curve Fitting] [Hyperlinear quantitative absorption spectrophotometry] [Appendix and Case Studies] [Peak Finding and Measurement] [iPeak] [iSignal] [Peak Fitters] [iFilter] [iPower] [List of downloadable software] [Interactive tools]
[Basic Properties of Derivatives] [Applications of Differentiation] [Peak detection] [Derivative Spectroscopy] [Trace Analysis] [The Importance of Smoothing Derivatives] [Video Demonstrations] [Spreadsheets] [Differentiation in Matlab and Octave] [Interactive tools] [Have a question? Email me]
The symbolic differentiation of functions is a topic that is introduced in all elementary Calculus courses. The numerical differentiation of digitized signals is an application of this concept that has many uses in analytical signal processing. The first derivative of a signal is the rate of change of y with x, that is, dy/dx, which is interpreted as the slope of the tangent to the signal at each point, as illustrated by the animation on the left (script). Assuming that the x-interval between adjacent points is constant, the simplest algorithm for computing a first derivative is:
1< j <n-1).
where X'j and Y'j are the X and Y values of the jth point of the derivative, n = number of points in the signal, and X is the difference between the X values of adjacent data points. A commonly used variation of this algorithm computes the average slope between three adjacent points:
< j <n-1).
This is called a central-difference method; its advantage is that it does not involve a shift in the x-axis position of the derivative. It's also possible to compute gap-segment derivatives in which the x-axis interval between the points in the above expressions is greater than one; for example, Yj-2 and Yj+2, or Yj-3 and Yj+3, etc. It turns out that this is equivalent to applying a moving-average (rectangular) smooth in addition to the derivative.
The second derivative is the derivative of the derivative: it is a measure of the curvature of the signal, that is, the rate of change of the slope of the signal. It can be calculated by applying the first derivative calculation twice in succession. The simplest algorithm for direct computation of the second derivative in one step is
< j <n-1).
Similarly, higher derivative orders can be
computed using the appropriate sequence of coefficients: for
example +1, -2, +2, -1 for the third derivative and +1, -4, +6,
-4, +1 for the 4th derivative, although these
derivatives can also be computed simply by taking successive
lower order derivatives. The first derivative can be interpreted
as the slope of the original at each point, and the
second derivative as the curvature, but beyond that we
have no single-word labels; each derivative is just the rate of
change of the one before it.
The Savitzky-Golay smooth
can also be used as a differentiation algorithm with the
appropriate choice of input arguments; it combines
differentiation and smoothing into one algorithm.
The accuracy of numerical
differentiation is demonstrated by the Matlab/Octave script GaussianDerivatives.m (graphic), which compares
the exact analytical expressions for the derivatives of
a Gaussian (readily
obtained from Wolfram Alpha) to the numerical values
obtained by the expressions above, demonstrating that the shape
and amplitude of the derivatives are an exact match as long as
the sampling interval is not too coarse. It also demonstrates
that the numerical nth derivative can be obtained
exactly by applying n successive first
differentiations. Ultimately, the numerical precision
limitation of the computer can be a limitation in some extreme cases.
Basic Properties of Derivative Signals
The figure on the left shows the results of the successive differentiation of a computer-generated Gaussian peak signal (click to see the full-sized figure). The signal in each of the four windows is the first derivative of the one before it; that is, Window 2 is the first derivative of Window 1, Window 3 is the first derivative of Window 2, Window 3 is the second derivative of Window 1, and so on. You can predict the shape of each signal by recalling that the derivative is simply the slope of the original signal: where a signal slopes up, its derivative is positive; where a signal slopes down, its derivative is negative; and where a signal has zero slope, its derivative is zero. (Matlab/Octave code for this figure.)
signal shown in Window 1 has an inflection point
(point where where the slope is maximum) at the center of the x
axis range. This corresponds to the maximum in its first
derivative (Window 2) and to the zero-crossing (point
where the signal crosses the x-axis going either from positive
to negative or vice versa) in the second derivative in
Window 3. This behavior can be useful for precisely locating the
inflection point in a sigmoid signal, by computing the location
of the zero-crossing in its second derivative. Similarly, the location of the maximum
in a peak-type signal can be computed precisely by computing the
location of the zero-crossing in its first derivative. Different
peak shapes have different derivatives shapes: the Matlab/Octave
demonstrates the first derivative forms of 16 different model
peak shapes (graphic). Any
smooth peak shape with a single maximum has sequential
derivatives that exhibit a series of alternating maxima and
minima, the total number of which is one more than the
derivative order. The
even-order derivatives have a maximum or a minimum at the peak
center, and the odd-order derivatives have a zero-crossing at the peak center (Graphic, Matlab/Octave code). You can
also see here that the numerical magnitude of the derivatives
(y-axis values) is much less than the original signal, because
derivatives are the differences between adjacent y
values, divided by the independent variable increment. (It's the
same reason the odometer in
your car usually displays a much larger number than the
speedometer; the speedometer is essentially the first derivative
of the odometer).
An important property of the
differentiation of peak-type signals is the effect of the peak width
on the amplitude of derivatives. The figure on the left shows
the results of the successive differentiation of two
computer-generated Gaussian bands (click to see the full-sized
figure). The two bands have the same amplitude (peak height) but
one of them is exactly twice the width of the other. As you can
see, the wider peak has the smaller derivative
amplitude, and the effect becomes more noticeable at higher
derivative orders. In general, it is found that that the
amplitude of the nth derivative of a
peak is inversely proportional to the nth
power of its width, for signals having the same shape and
amplitude. Thus differentiation in effect discriminates against
wider peaks and the higher the order of differentiation the
greater the discrimination. This behavior can be useful in
quantitative analytical applications for detecting peaks that
are superimposed on and obscured by stronger but broader
background peaks. (Matlab/Octave code
for this figure). The amplitude of a derivative of a peak also
depends on the shape of the peak and is directly
proportional to its peak height. Gaussian and Lorentzian
peak shapes have slightly different first and second derivative
shapes and amplitudes. The amplitude of the nth derivative
of a Gaussian peak of height H and width W can be estimated by
the empirical equation
H*(10^(0.027*n^2+n*0.45-0.31)).*W^(-n), where W is the full
width at half maximum (FWHM) measured in the number of x,y data
Although differentiation changes the shape of peak-type signals drastically, a periodic signal like a sine wave signal behaves very differently. The derivative of a sine wave of frequency f is a phase-shifted sine wave, or cosine wave, of the same frequency and with an amplitude that is proportional to f, as can be demonstrated in Wolfram Alpha. The derivative of a periodic signal containing several sine components of different frequency will still contain those same frequencies, but with altered amplitudes and phases. For this reason, when a music or speech signal is differentiated, the music or speech is still completely recognizable, but with the high frequencies increased in amplitude compared to the low frequencies, and as a result, it sounds "thin" or "tinny". See this demonstration for an example.
Applications of Differentiation
A simple example of the application of differentiation of experimental signals is shown in the figure below. This signal is typical of the type of signal recorded in amperometric titrations and some kinds of thermal analysis and kinetic experiments: a series of straight line segments of different slope. The objective is to determine how many segments there are, where the breaks between then fall, and the slopes of each segment. This is difficult to do from the raw data, because the slope differences are small and the resolution of the computer screen display is limiting. The task is much simpler if the first derivative (slope) of the signal is calculated (below right). Each segment is now clearly seen as a separate step whose height (y-axis value) is the slope. The y-axis now takes on the units of dy/dx. Note that in this example the steps in the derivative signal are not completely flat, indicating that the line segments in the original signal were not perfectly straight. This is most likely due to random noise in the original signal. Although this noise was not particularly evident in the original signal, it is more noticeable in the derivative.
The signal on the left seems to be a more-or-less straight line, but its numerically calculated derivative (dx/dy), plotted on the right, shows that the line actually has several approximately straight-line segments with distinctly different slopes and with well-defined breaks between each segment.
It is commonly observed that differentiation degrades signal-to-noise ratio, unless the differentiation algorithm includes smoothing that is carefully optimized for each application. Numerical algorithms for differentiation are as numerous as for smoothing and must be carefully chosen to control signal-to-noise degradation.
A classic use of second differentiation in chemical analysis is in the location of endpoints in potentiometric titration. In most titrations, the titration curve has a sigmoidal shape and the endpoint is indicated by the inflection point, the point where the slope is maximum and the curvature is zero. The first derivative of the titration curve will therefore exhibit a maximum at the inflection point, and the second derivative will exhibit a zero-crossing at that point. Maxima and zero crossings are usually much easier to locate precisely than inflection points.
The signal on the left is the pH titration curve of a very weak acid with a strong base, with volume in mL on the X-axis and pH on the Y-axis. The endpoint is the point of greatest slope; this is also an inflection point, where the curvature of the signal is zero. With a weak acid such as this, it is difficult to locate this point precisely from the original titration curve. The endpoint is much more easily located in the second derivative, shown on the right, as the zero crossing.
The figure above shows a pH titration curve of a
very weak acid with a strong base, with volume in mL on the
X-axis and pH on the Y-axis. The volumetric equivalence point
(the "theoretical" endpoint) is 20 mL. The endpoint is the point
of greatest slope; this is also an inflection point, where the
curvature of the signal is zero. With a weak acid such as this,
it is difficult to locate this point precisely from the original
titration curve. The second derivative of the curve is shown in
Window 2 on the right. The zero crossing of the second
derivative corresponds to the endpoint and is much more
precisely measurable. Note that in the second derivative plot,
both the x-axis and the y-axis scales have been expanded to show
the zero crossing point more clearly. The dotted lines show that
the zero crossing falls at about 19.4 mL, close to the
theoretical value of 20 mL.
Derivatives can also be used as a simple way to
detect unexpected asymmetry in otherwise symmetrical peaks. For
example, a pure Gaussian peaks is symmetrical but, if subjected
broadening, can become asymmetrical. If the degree of
broadening is small, it can be difficult to detect visually;
that is where differentiation can help. DerivativeEMGDemo.m (graphic) shows the 1st
through 5th derivatives of a slightly exponentially broadened
Gaussian (EMG); of those derivatives, the second clearly shows unequal
positive peaks that would be expected to be equal for a
purely symmetrical peak. The higher derivatives offer no clear
advantage and are more susceptible to white noise in the signal.
For another example, if a Gaussian peak is
heavily overlapped by a smaller peak, the result is usually
asymmetrical. The script DerivativePeakOverlapDemo
(graphic) shows the
1st through 5th derivatives of two
overlapping Gaussians where the second peak is so small and so
close that it's impossible to discern visually, but again the
second derivative shows the asymmetry clearly by comparing the
heights of the two positive peaks. DerivativePeakOverlap.m
detects the minimum extent of peak overlap by the first
and second derivatives, looking for the point at which two peaks
are visible; for each trial separation, it prints out the
separation, resolution, and the number of peaks detected in the
first and second derivatives.
Another common use of differentiation is in
the detection of peaks in a signal. It's clear from the basic
properties described in the previous section that the first
derivative of a peak has a downward-going zero-crossing at the
peak maximum, which can be used to locate the x-value of the
peak, as shown on the right (script).
If there is no
noise in the signal, then any data point that has lower
values on both sides of it will be a peak maximum. But there is
always at least a little noise in real experimental signals, and
that will cause many false zero-crossings simply due to the
noise. To avoid this problem, one popular
technique smooths the first derivative of the signal
first, before looking for downward-going zero-crossings, and
then takes only those zero crossings whose slope exceeds a
certain predetermined minimum (called the "slope threshold") at
a point where the original signal amplitude exceeds a certain
minimum (called the "amplitude threshold"). By carefully
adjusting the smooth width, slope threshold, and amplitude
threshold, it is possible to detect only the desired peaks over
a wide range of peak widths and ignore peaks that are too small,
too wide, or too narrow. Moreover, because smoothing can distort peak
signals, reducing peak heights, and increasing peak
widths, this technique can be extended to measure the position,
height, and width of each peak by least-squares
curve-fitting of a segment of original unsmoothed signal near
the top of the peak (where the signal-to-noise ratio is
usually the best). Thus, even if heavy smoothing is necessary to
provide reliable discrimination against noise peaks, the peak
parameters extracted by curve fitting are not distorted and the
effect of random noise in the signal is reduced by curve fitting
over multiple data points in the peak. This technique has been
implemented in Matlab/Octave
and in spreadsheets.
In spectroscopy, the differentiation of spectra
is a widely used technique, particularly in infra-red,
spectrophotometry, referred to as derivative
spectroscopy. Derivative methods have been used in
analytical spectroscopy for three main purposes:
(a) spectral discrimination, as a qualitative fingerprinting technique to accentuate small structural differences between nearly identical spectra;
(b) spectral resolution enhancement, as a technique for increasing the apparent resolution of overlapping spectral bands in order to more easily determine the number of bands and their wavelengths;
(c) quantitative analysis, as a technique for the correction for irrelevant background absorption and as a way to facilitate multicomponent analysis. (Because differentiation is a linear technique, the amplitude of a derivative is proportional to the amplitude of the original signal, which allows quantitative analysis applications employing any of the standard calibration techniques). Most commercial spectrophotometers now have built-in derivative capability. Some instruments are designed to measure the spectral derivatives optically, by means of dual wavelength or wavelength modulation designs.
Because of the fact that the amplitude of the nth derivative of a peak-shaped signal is inversely proportional to the nth power of the width of the peak, differentiation may be employed as a general way to discriminate against broad spectral features in favor of narrow components. This is the basis for the application of differentiation as a method of correction for background signals in quantitative spectrophotometric analysis. Very often in the practical applications of spectrophotometry to the analysis of complex samples, the spectral bands of the analyte (i.e. the compound to be measured) are superimposed on a broad, gradually curved background. Background of this type can be reduced by differentiation.
This is illustrated by the figure above, which shows a simulated UV spectrum (absorbance vs wavelength in nm), with the green curve representing the spectrum of the pure analyte and the red line representing the spectrum of a mixture containing the analyte plus other compounds that give rise to the large sloping background absorption. The first derivatives of these two signals are shown in the center; you can see that the difference between the pure analyte spectrum (green) and the mixture spectrum (red) is reduced. This effect is considerably enhanced in the second derivative, shown on the right. In this case the spectra of the pure analyte and of the mixture are almost identical. In order for this technique to work, it is necessary that the background absorption be broader (that is, have lower curvature) than the analyte spectral peak, but this turns out to be a rather common situation. Because of their greater discrimination against broad background, second (and sometimes even higher-order) derivatives are often used for such purposes. See DerivativeDemo.m for a Matlab/Octave demonstration of this application.
It is sometimes (mistakenly) said that differentiation "increases the sensitivity" of analysis. You can see how it would be tempting to say something like that by inspecting the three figures above; it does seems that the signal amplitude of the derivatives is greater than that of the original analyte signal (at least graphically). However, it is not valid to compare the amplitudes of signals and their derivatives because they have different units. The units of the original spectrum are absorbance; the units of the first derivative are absorbance per nm, and the units of the second derivative are absorbance per nm2. You can't compare absorbance to absorbance per nm any more than you can compare miles to miles per hour. (It's meaningless, for instance, to say that a speed of 30 miles per hour is greater than a distance of 20 miles.) You can, however, compare the signal-to-background ratio and the signal-to-noise ratio. For instance, in the above example, it would be valid to say that the signal-to-background ratio is increased in the derivatives.
Loosely speaking, the opposite of differentiation is integration, so if you are given a first derivative of a signal, you might expect to regenerate the original (zeroth derivative) by integration. However, there is a catch; the constant term in original signal (like a flat baseline) is completely lost in differentiation; integration can not restore it. So strictly speaking, differentiation represents a net loss of information, and therefore differentiation is used only in situations where the constant term in the original signal is not of interest.
There are several ways to measure the amplitude
of a derivative spectrum for quantitative analysis: the absolute
value of the derivative at a specific wavelength, the value of a
specific feature (such as a maximum), or the difference between
a maximum and a minimum. Another widely-used technique is the
zero-crossing measurement - taking readings derivative amplitude
at the wavelength where an interfering peak crosses the zero on
the y (amplitude) axis. In all these cases, it's important to
measure the standards and the unknown samples in exactly the
same way. Also, because the amplitude of a derivative of a
peak depends strongly on its width, it's important to control
environmental factors that might change spectral peak width
subtlety, such as temperature.
One of the widest uses of the derivative signal
processing technique in practical analytical work is in the
measurement of small amounts of substances in the presence of
large amounts of potentially interfering materials. In such
applications it is common that the analytical signals are weak,
noisy, and superimposed on large background signals. Measurement
precision is often degraded by sample-to-sample baseline shifts
due to non-specific broadband interfering absorption,
non-reproducible cuvette (sample cell) positioning, dirt or
fingerprints on the cuvette walls, imperfect cuvette
transmission matching, and solution turbidity. Baseline shifts
from these sources are usually either wavelength-independent
(light blockage caused by bubbles or large suspended particles)
or exhibit a weak wavelength dependence (small-particle
turbidity). Therefore it can be expected that differentiation
will in general help to discriminate relevant absorption from
these sources of baseline shift. An obvious benefit of the
suppression of broad background by differentiation is that variations
in the background amplitude from sample to sample are also
reduced. This can result in improved precision or measurement in
many instances, especially when the analyte signal is small
compared to the background and if there is a lot of uncontrolled
variability in the background. An example of the improved
ability to detect trace component in the presence of strong
background interference is shown in this figure:
The spectrum on the left shows a weak
shoulder near the center due to a small concentration of the
substance that is to be measured (e.g. the active ingredient
in a pharmaceutical preparation). It is difficult to measure
the intensity of this peak because it is obscured by the
strong background caused by other substances in the sample.
The fourth derivative of this spectrum is shown on the
right. The background has been almost completely suppressed
and the analyte peak now stands out clearly, facilitating
The spectrum on the left shows a weak shoulder near the center due to the analyte. The signal-to-noise ratio is very good in this spectrum, but in spite of that the broad, sloping background obscures the peak and makes quantitative measurement very difficult. The fourth derivative of this spectrum is shown on the right. The background has been almost completely suppressed and the analyte peak now stands out clearly, facilitating measurement. An even more dramatic case is shown below. This is essentially the same spectrum as in the figure above, except that the concentration of the analyte is ten times lower. The question is: is there a detectable amount of analyte in this spectrum? This is quite impossible to say from the normal spectrum, but inspection of the fourth derivative (right) shows that the answer is yes. Some noise is clearly evident here, but nevertheless the signal-to-noise ratio is sufficiently good for a reasonable quantitative measurement.
Similar to the previous figure, but in the case the peak is ten times lower - so weak that it can not even be seen in the spectrum on the left. The fourth derivative (right) shows that a peak is still there, but much reduced in amplitude (note the smaller y-axis scale).
This use of signal differentiation has become
widely used in quantitative
spectroscopy, particularly for quality control in the pharmaceutical
industry. In that application the analyte would typically
be the active ingredient in a pharmaceutical preparation and the
background interferences might arise from the presence of
fillers, emulsifiers, flavoring or coloring agents, buffers,
stabilizers, or other excipients. Of course, in trace
analysis applications, care must be taken to optimize signal-to-noise
ratio of the instrument as much as possible.
Although it will eventually be shown that more
advanced techniques such as curve fitting can also
perform many of these quantitative measurement tasks quite well,
the derivative techniques have the advantage of conceptual and
mathematical simplicity and an easily-understood graphical way
of presenting presenting data.
and Noise: The Importance of Smoothing
It is often said that
"differentiation increases the noise". That is true, but it is
not the main problem. In fact, computing the unsmoothed first
derivative of a set of random numbers increases its standard deviation by only the
square root of 2, simply due to the usual propagation of errors of the
sum or difference between two numbers. As an example, the
standard deviation (std) of the numbers generated by the Matlab/Octave
randn() function is 1.0 and its first
derivative std(deriv1(randn(size(1:10000)))) equals about
1.4. But even a little bit of smoothing applied to the
derivative will reduce this standard deviation greatly, e.g. std(fastsmooth(deriv1(randn(size(1:10000))),2,3)) equals about 0.4. More important is the fact that
the signal-to-noise ratio of an unsmoothed derivative
is almost always much lower (poorer) than that of the original
signal, mainly because the numerical amplitude of the
derivative is usually much smaller (as you can see for
yourself in all the examples on this page). But smoothing is always used in any
practical application to control this problem; with
optimal smoothing, the signal-to-noise of a derivative
can actually be greater than the unsmoothed
original. For the successful
application of differentiation in quantitative analytical
applications, it is essential to use differentiation in
combination with sufficient smoothing, in order to optimize the
signal-to-noise ratio. This is illustrated
in the figure on the left. (Matlab/Octave
code for this figure.) Window 1 shows a Gaussian band with
a small amount of added white noise. Windows 2, 3, and 4, show
the first derivative of that signal with increasing smooth
widths. As you can see, without sufficient smoothing, the
signal-to-noise ratio of the derivative can be substantially
poorer than the original signal. However, with adequate
amounts of smoothing, the signal-to-noise ratio of the smoothed
derivative can be better than that of the unsmoothed original.
This effect is even more striking in the second derivative, as
shown on the right (Matlab/Octave code
for this figure). In this case, the signal-to-noise ratio of the
unsmoothed second derivative (Window 2) is so poor you can not
even see the signal visually, but the smoothed second derivative
looks fine. Differentiation does not actually add noise to
the signal; if there were no noise at all in the original
signal, then the derivatives would also have no noise
(exception: see Appendix V).
What is particularly interesting about the noise
in these derivative signals, however, is their "color". This noise
is not white; rather, it is blue - that is, it
has much more power at high frequencies than white noise. The
consequence of this is that the noise in differentiated signal
is easily reduced greatly by smoothing,
as demonstrated above.
It makes no difference whether the smooth
operation is applied before or after the differentiation. What
is important, however, is the nature of the smooth, its smooth
ratio (ratio of the smooth width to the width of the original
peak), and the number of times the signal is smoothed. The
optimum values of smooth ratio for derivative signals is
approximately 0.5 to 1.0. For a first derivative, two applications
of a simple rectangular smooth (or one application of a
triangular smooth) is adequate. For a second derivative,
three applications of a simple rectangular smooth or two
applications of a triangular smooth is adequate. The general
rule is: for the nth derivative, use
at least n+1 applications of a rectangular smooth. (The Matlab
signal processing program iSignal automatically
provides the desired type of smooth for each derivative order).
A quantitative comparison of smoothing the
derivatives is performed by the script MultiPeakDerivativeOptimization.m,
which analyzes the second derivatives of Gaussian peaks smoothed
by moving average, triangular, Gaussian, and Suavity-Golay
If the peak widths vary substantially across the
signal recording - for example, if the peaks get regularly wider
as the x-value increases - then it may be helpful to use an adaptive segmented
smooth, which makes the smooth width vary across the
Smoothing derivative signals usually results in a substantial attenuation of the derivative amplitude; in the figure on the right above, the amplitude of the most heavily smoothed derivative (in Window 4) is much less than its less-smoothed version (Window 3). However, this won't be a problem in quantitative analysis applications, as long as the standard (analytical) curve is prepared using the exact same derivative, smoothing, and measurement procedure as is applied to the unknown samples. Because differentiation and smoothing are both linear techniques, the amplitude of a smoothed derivative is exactly proportional to the amplitude of the original signal, which allows quantitative analysis applications employing any of the standard calibration techniques. As long as you apply the same signal-processing techniques to the standards as well as to the samples, everything works.
Because of the different kinds and degrees of smoothing that might be incorporated into the computation of digital differentiation of experimental signals, it's difficult to compare the results of different instruments and experiments unless the details of these computations are known. In commercial instruments and software packages, these details may well be hidden. However, if you can obtain both the original (zeroth derivative) signal, as well as the derivative and/or smoothed version from the same instrument or software package, then the technique of Fourier deconvolution, which will be discussed later, can be used to discover and duplicate the underlying hidden computations.
Interestingly, neglecting to smooth a derivative was ultimately responsible for the failure of the first spacecraft of NASA's Mariner program on July 22, 1962, which was reported in InfoWorld's "11 infamous software bugs". In his 1968 book "The Promise of Space", Arthur C. Clarke described the mission as "wrecked by the most expensive hyphen in history." The "hyphen" was actually superscript bar over the symbol for velocity (the first derivative of position), handwritten in a notebook. An overbar conventionally signifies an averaging or smoothing function, so the formula should have calculated the smoothed value of the time derivative of position. Without the smoothing function, even minor variations would cause its derivative to be very noisy and to trigger the corrective boosters to kick in prematurely, causing the rocket's flight to become unstable.
The first 13-second, 1.5 MByte video (SmoothDerivative2.wmv ) demonstrates the huge signal-to-noise ratio improvements that are possible when smoothing derivative signals, in this case a 4th derivative.
The second video, 17-second, 1.1 MByte, (DerivativeBackground2.wmv )
demonstrates the measurement of a weak peak buried in a strong
sloping background. At the beginning of this brief video, the
amplitude (Amp) of the peak is varied between 0 and 0.14, but
the background is so strong that the peak, located at x = 500,
is hardly visible. Then the 4th derivative (Order=4) is computed
and the scale expansion (Scale) is increased, with a smooth
width (Smooth) of 88. Finally, the amplitude (Amp) of the peak
is varied again over the same range, but now the changes in the
signal are now quite noticeable and easily measured. (These
demonstrations were created in Matlab 6.5. If you have access to
that software, you may download a set of Matlab Interactive
Derivative m-files (15 Kbytes), InteractiveDerivative.zip
so that you can experiment with the variables at will and try
out this technique on your own signals).
Differentiation in Spreadsheets
Differentiation operations such as described
above can readily be performed in spreadsheets such as Excel or
OpenOffice Calc. Both the derivative and the required smoothing
operations can be performed by the shift-and-multiply method
described in the section on smoothing.
In principle, it is possible to combine any degree of
differentiation and smoothing into one set of shift-and-multiply
illustrated here), but it's more flexible and easier to
adjust if you compute the derivatives and each stage of
smoothing separately in successive columns. This is illustrated
for OpenOffice Calc and DerivativeSmoothing.xls
for Excel (screen image),
which smooths the data and computes the first derivative of Y (column B) with
respect to X (column A), then applies that
smoothing and differentiation process successively to compute
the smoothed second and third derivatives. The same smoothing
coefficients (in row 5, columns K through AA)
are applied successively for each stage of differentiation; you
can enter any set of numbers here (preferably symmetrical about
the center number in column S). You can type or paste
your own data into column A and B (X and Y),
rows 8 to 263.
demonstrates the dramatic effect of smoothing on the
signal-to-noise ratio of derivatives on a noisy signal. It uses
the same signal as DerivativeSmoothing.xls,
but adds simulated white noise to the Y data. You can control
the amount of added noise (cell D5).
Another example of a derivative application is the spreadsheet SecondDerivativeXY2.xlsx (graphic), which demonstrates locating and measuring changes in the second derivative (a measure of curvature or acceleration) of a time-changing signal. This spreadsheet shows the apparent increase in noise caused by differentiation and the extent to which the noise can be reduced by smoothing (in this case by two passes of a 5-point triangular smooth). The smoothed second derivative shows a large peak the point at which the acceleration changes (at x=30) and plateaus on either side showing the magnitude of the acceleration before and after the change (y=2 and 4, respectively).
Differentiation in Matlab and Octave
| Differentiation functions such as described above
can easily be created in Matlab or Octave. Some
simple derivative functions for equally-spaced time series
data: deriv, a first derivative
using the 2-point central-difference method, deriv1, an unsmoothed first
derivative using adjacent differences, deriv2, a second derivative using
the 3-point central-difference method, a third derivative
deriv3 using a 4-point formula, and
deriv4, a 4th derivative using a
5-point formula. Each of these is a simple Matlab function
of the form d=deriv(y);
the input argument is a signal vector "y", and the
differentiated signal is returned as the vector "d". For data that are
equally-spaced on the independent variable (x) axis, there
are versions of the first and second derivative
functions, derivxy and secderivxy, that take two input
arguments (x,y), where x
and y are vectors
containing the independent and dependent variables. Click
on these links to inspect the code, or right-click to
download for use within Matlab. All of these are unsmoothed
derivative that usually must be smoothed either after
or before differentiation.
Smoothed derivatives. Since smoothing almost always goes along with differentiation, the function SmoothDerivative.m combines the two. The syntax is function Processed=SmoothedDerivative(x, y, DerivativeOrder, w, type, ends) where
'DerivativeOrder' determines the derivative order (0, 1, 2, 3, 4, 5), 'w' is the smooth width, 'type' determines the smooth mode:
If type=0, the signal is not smoothed.
If type=1, rectangular (sliding-average or boxcar)
If type=2, triangular (2 passes of sliding-average)
If type=3, pseudo-Gaussian (3 passes of sliding-average)
If type=4, Savitzky-Golay smooth
and 'ends' controls how the "ends" of the signal (the first w/2 points and the last w/2 points) are handled
If ends=0, the ends are zeroed
If ends=1, the ends are smoothed with progressively smaller smooths the closer to the end.
Peak detection. The simplest code to find peaks in x,y data sets simply looks for every y value that has lower y values on both sides (allpeaks.m). A alternative approach is to use the first derivative to find all the maxima by locating the points of zero-crossing, that is, the points at which the first derivative "d" (computed by derivxy.m) passes from positive to negative. In this example, the "sign" function is a built-in function that returns 1 if the element is greater than zero, 0 if it equals zero and -1 if it is less than zero. The routine prints out the value of x and y at each zero-crossing:
If the data are noisy, many false zero crossings will be reported; smoothing the data will reduce that. If the data are sparsely sampled, a more accurate value for the peak position (x-axis value at the zero crossing) can be obtained by interpolating between the point before and the point after the zero-crossing using the Matlab/Octave "interp1" function: interp1([d(j) d(j+1)],[x(j) x(j+1)],0)
ProcessSignal.m, a Matlab/Octave command-line function that performs smoothing and differentiation on the time-series data set x,y (column or row vectors). Type "help ProcessSignal". Returns the processed signal as a vector that has the same shape as x, regardless of the shape of y. The syntax is Processed=ProcessSignal(x, y, DerivativeMode, w, type, ends, Sharpen, factor1, factor2, SlewRate, MedianWidth)
iSignal (shown above) is an interactive multipurpose signal processing function for Matlab that includes differentiation and smoothing for time-series signals, up to the 5th derivative, automatically including the required type of smoothing. Simple keystrokes allow you to adjust the smoothing parameters (smooth type, width, and ends treatment) while observing the effect on your signal dynamically. In the example shown above, a series of three peaks at x=100, 250, and 400, with heights in the ratio 1:2:3, are buried in a strong curved background; the smoothed second and fourth derivatives are computed to suppress that background. View the code here or download the ZIP file with sample data for testing. (Version 2 of iSignal, November 2011, computes derivatives with respect to the x-axis vector, correcting for non-uniform x-axis intervals). Unfortunately, iSignal does not currently work in Octave.
These statements generate the 4th derivative of a Gaussian peak and display it in iSignal. You'll need to download isignal.m, gaussian.m, and deriv4.m.
The signal is mostly blue noise (because of the differentiated white noise) unless you smooth it considerately. Use the A and Z keys to increase and decrease the smooth width and the S key to cycle through the available smooth types. Hint: use the Gaussian smooth and keep increasing the smooth width.
The script "iSignalDeltaTest" demonstrates the frequency response of the smoothing and differentiation functions of iSignal by a applying them to a delta function. Change the smooth type, smooth width, and derivative order and see how the power spectrum changes.
differentiation in Matlab is discussed in Appendix Y.