Computes the theoretical signal-to-noise ratio for a UV-visible absorption specrophotometer with a continuum source (e.g. tungsten incandescent lamp) modeled as a blackbody, dispersive monochromator, and a photomultiplier detector, given the molar absorptivity of the analyte and the solvent at the measured wavelength. Includes unabsorbed stray light, source flicker, photon, and thermionic emission noise. Note that this simulation predicts only a theoretical maximum signal-to-noise ratio limited by photon and detector noise; in practice, the limits to the measurement of small absorbances are more likely set by low-frequency flicker noise and sample-to-sample baseline variations, caused by light source instability, vibration, sample turbulence, light scattering by suspended particles, dust, bubbles, fingerprints on the cuvettes, etc. Most of those noises can be reduced or eliminated by using double-beam, dual-wavelength, derivative spectroscopy, wavelength modulation and photodiode array instruments, coupled with suitable multicomponent software.
Assumptions: The Beer-Lambert law is observed, except for the presence of unabsorbed stray light. Double-beam design measures both sample and reference beams, and both beams are subject to source flicker, photon, and thermionic emission noise. Source flicker, photon, and thermionic emission noises are random and uncorrelated and therefore add quadratically. Source flicker is only that portion of the source flicker not compensated by the double-beam operation. The detector is linear. Quantum efficiency of photocathode is independent of light level and detector current. Spectral response of the detector is much wider that the spectral bandpass of the monochromator.
View Equations (.pdf)
Download spreadsheet in OpenOffice format (.ods)
Download spreadsheet in Excel format (.xls)
Other related simulations:
Dual Wavelength Spectrophotometer
Effect of Slit Width on Signal-to-Noise Ratio in Absorption Spectroscopy
[Return to Index]
Note: to run the OpenOffice (.ods) spreadsheets, you have to first download the OpenOffice installer (download from OpenOffice), then install it (by double-clicking on the installer file that you just downloaded), and then download my spreadsheets from this page. Once OpenOffice is installed, you can run my spreadsheets just by double-clicking on them. Note 1: Don't use version 3.1. There is a bug in OpenOffice 3.1 that causes bad x-axis scaling on some of my graphs. The problems does not occur in version 3.0 or in the most recent version 3.2. Note 2: Downloading these files with Interent Explorer will change the file types from ".ods" to ".zip"; you will have to edit the file names and change the extensions back to ".ods" for them to work properly. This problem does not occur in Firefox or in Chrome.