Computes the theoretical signal-to-noise ratio for a fluorescence specrometer with a continuum source (e.g. Xenon arc lamp) modeled as a blackbody, dispersive emission and excitation monochromators, and a photomultiplier detector, given the molar absorptivity and fluorescence quantum yield of the analyte at the excitation wavelength. Note that this simulation predicts only a theoretical maximum signal-to-noise ratio limited by detector noise and by the photon noise of the analyte's fluorescence emission. In practice, the measurement of small fluorescence intensities is likely to be limited by photon noise from spectral interferences, background fluorescence from the solvent, or the contamination on the sample cuvette. Even if the systematic error caused by background fluorescence interference is eliminated or compensated (for example by using synchronous scanning, wavelength modulation, derivative spectroscopy, or advanced chemometric techniques such as rank annihilation), the photon noise from the background remains and sets an ultimate limit to signal-to-noise ratio performance.
Assumptions: Room-temperature prompt fluorescence of a stable fluorophor in aqueous solution with right-angle geometry in a standard cuvette, measured with a dispersive spectrofluorometer; no background fluorescence from the solvent; scattering is resolved from the fluorescence; self-absorption is ignored in this version; sample cell is larger than entrance slit height and width; fluorescence emission is isotropic and completely fills the emission monochromator solid angle. Spectral response of the detector is much wider that the spectral bandpass of the monochromator.
Note: The initial values for molar absorptivity (5600) and quantum efficiency (0.53) are for quinine sulfate in aqueous solution excited at 350 nm.
Download spreadsheet in OpenOffice format (.ods)
Download spreadsheet in Excel format (.xls)
Note: to run the OpenOffice
(.ods) spreadsheets, you have to first download the OpenOffice
from OpenOffice), then install it (by double-clicking on the
installer file that you just downloaded), and then download my
spreadsheets from this page. Once OpenOffice is installed,
you can run my spreadsheets just by double-clicking on them.
Note 1: Don't use version 3.1. There
is a bug in OpenOffice 3.1 that causes bad x-axis scaling on some
of my graphs. The problems does not occur in version 3.0 or
in the most recent version 3.2. Note 2: Downloading these files with Interent Explorer will change
the file types from ".ods" to ".zip"; you will have to edit the
file names and change the extensions back to ".ods" for them to
work properly. This problem does not occur in Firefox or in Chrome.
Scanning Fluorescence Spectrometer
Signal-to-noise ratio of absorption spectrophotometry
[Return to Index of Simulations]